How to install and debug Weighing Module ？
As long as people are still "haggling", the scale is very useful. In special occasions, the general electronic scale can not be used. Is there a way to avoid the fixed electronic scale and only use it sensor What about it? Let's talk about how to install and debug the weighing module
Taking a typical chemical reaction kettle as an example, three pressure weighing modules installed at the bottom of the reactor transmit the weight signal of the reactor body to the junction box. Each weighing module signal is paralleled in the junction box, and the angle difference can be adjusted according to the load condition of each weighing module, and the weight signal after collection is sent to the weighing end or weight Transmitter 。 The weighing terminal or weight transmitter processes the weight signal into corresponding weight data.
FW / GW weighting module installation
Stainless steel material is used to effectively resist corrosion of corrosive substances. Fixed, floating and semi floating structures can be used in various harsh conditions.
CW / uW weighting module I installation
In the installation mode of four dynamic load modules, two modules are limited on three sides, and the other two modules are limited at the end.
Weighing module applications
Application of weighing module
According to the force characteristics of the weighing module in practical application, the weighing module is divided into two types: static load weighing module and dynamic load weighing module. The static load weighing module is suitable for the situation of only vertical force. Such as reactor, tank, hopper weight detection. The dynamic load module is suitable for the occasion of obvious force. Such as platform, production line, conveyor belt weight detection.
Weighing applied in
1. Weight monitoring
2. Quantitative control
3. Flow level detection
4. Bulk aggregate s
5. Force measurement
6. Weight / sorting / counting
7. Flow feeding
Principles of weighing module
1. For non professional users, direct use The weighing sensor It's very difficult.
2. The weighing module is a new type of weighing element. It combines the weighing sensor, load transfer device and installation connector together, which can be easily connected with mechanical devices of various shapes, such as raceway, platform, vertical tank, tank, hopper, etc.
3. The system with weighing module can not only ensure the high precision and long-term stability of the weighing sensor, but also solve the weighing error caused by improper installation.
4. The measurement accuracy of weighing module is better than 5 / 10000.